Depth of anesthesia and sedation monitor
Intensive care during postoperative period
The device can work with any type of anesthesia machines
The device analyzes patient`s EEG in time and frequency domains and defines brain activity index (AI), which reflects patient`s depth of anesthesia and level of sedation.
|AI - Brain activity index|| |
Indicates patient's depth of anesthesia and sedation level.
Depth of anesthesia evaluation is based on analisys of electroencephalogram (EEG).
Algorithm of EEG analisys keeps information about typical signs of influence of different groups of anesthetics - types of EEG suppression
|SR - EEG signal suppresion rate||Reflects the suppression of EEG activity and determines level of the further deepening of anesthesia|
|SQI - EEG signal quality index||Indicates the influence of impedance of cable electrodes, artifacts, high-frequency noise, power mains disturbances and decreases linearly from 100% to 0%|
|EMG - Electromyographic component level||Electrical activity of facial muscles|
|Patient age groups||Adults, children 10 years old and older|
|Display||4.3" color, touch screen|
|Measurement method||Data processing of one electroencephalogram lead|
|Power supply|| |
Built-in battery 2 hours of operation
100–250 V, 50/60 Hz
|Scheme of applying of electrodes|| |
Derivation points F7-FZ
F8-FZ (a frontotemporal part)
|Displaying of registered parameters|| |
|Displaying of digital parameters|| |
Brain activity index (AI)
Signal suppression rate (SR)
Signal quality index (SQI)
Electromyographic component level (EMG)
Impedance of electrodes (Z1-Z3)
Low level of alarm on AI
High level of alarm on AI
Visual and sound alarm system
Technical alarms: electrode is disconnected, HF noise, power line disturbances, multiple artifacts, low battery
24 hours of graphical trends
|Mount||Universal mounting on a horizontal surface, in vertical and horizontal planes|
Features and advantages
|Universality||Operates with inhalation and intravenous anesthetics|
|Low cost in operation||Kit from 3 typical ECG-electrodes is used for measurement|
|Simplicity and safety for patient||3 ECG-electrodes are placed on the fronto-temporal lobes of the patient|
Using the device allows:
The information is for reference only. The list of features is not complete. For the detailed information please contact the manufacturer.
Not a public offer.